# Voltage Transducer | Selecting and Using Transducers Series: For the Measurement of Electric, Power, Voltage, and Current

Print

**What is a voltage transducer? **Voltage transducers provide a DC (direct current) or voltage output directly proportional to the AC input voltage. AC voltage transducers typically have a transformer input to isolate the transducer from the voltage input.

Following the transformer are the electronics.

There are two types of AC voltage transducers.

- Absolute average measuring, RMS calibrated (or mean value measuring, RMS calibrated).

These inexpensive transducers simply convert the AC input to DC and have the output calibrated to represent the root mean square (RMS) value for sine wave input. This type is very adequate for situations in which the voltage wave shape is not distorted. Any odd harmonic or discontinuity will introduce large error. Use the true RMS measuring type when distortion of a sine wave is present such as Variable Frequency Drives and SCR controlled loads.

- True RMS (Root Mean Square) measuring.

These transducers calculate the RMS value of the voltage input and provide a DC output directly proportional to the effective value of the voltage input. This type should be used whenever the voltage is distorted. Transducer models are available for nominal input voltages of 69, 120, 240, and 480 volts. These typically have a measuring range of 0 to 125% of the nominal input rating. Thus a 120-volt model has a range of 0 to 150 volts. For voltage input higher than 600 volts, one should use a potential transformer.

Interested in reading the entire Selecting and Using Transducers for the Measurement of Electric, Power, Voltage, and Current document? (Click here)

Need more help selecting the best and most accurate power measurement tool for your project? Call Ohio Semitronics, Inc. today, 1-800-537-6732

sales@ohiosemitronics.com